Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||18 August 2016|
|PDF File Size:||3.90 Mb|
|ePub File Size:||3.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Message 5 of manjal. Should I be looking at doing it with LabView or something else? It’s 84MB, so I hope you have a fast link.
Message 6 of 6.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. Message 3 of 6.
Good luck with it Hp4119a 1 of 6. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
I suspect that this difference might be causing my problems. I am able to communicate with the device through the GPIB interface, and am also able to retrieve information. IIRC, the sweep parameters were something like this: In particular, after setting up and initiating the instrument, I am initiating the measurement and reading manuall as follows: I think the would probably take the old commands, but the ALSO has new commands.
If not, what would creating a driver for it entail? Would you mind posting the code you used on the A?
Message 4 of 6. Message 2 of 6.
4191A RF IMPEDANCE ANL
Matt – you’re right, the commands for the A are completely different from the – probably because of its age. I have worked with hp4911a A and, as I recall, the commands are nothing like any other instrument I have seen.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve hp4191w seen before that I assumed it was referring to something different. Sorry, that last line of code there is supposed to read: Thanks for your help guys.
Agilent / HP 4191A RF Impedance Analyzer
You really need the manual. In the code above, MHz is an arbitrary frequency. I have not worked with that instrument, but the Agilent web site has the operating hpp4191a service manual here. Accepted by topic author Matt C.
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. When I gave it a try, ph4191a, it worked fine.
Has anyone done anything on interfacing with the A since Matt C’s last post? Guess you never know until you try! The forums emoticons took over for a bit there. Most Active Software Boards: